|FA23873||Can I access the diagnostics of an OTB island using the Advantys software?|
No, there are no diagnostic functions for an OTB island in the Advantys software.
The OTB NIM modules manage a limited number of online operations,...
|FA22755||How can I perform the "start test" function on an XUSLA light curtain monitored by an XPSMP configurable module?|
Test mode function: This function is neither mandatory nor critical since the XUSLx light curtain has a continuous self-test function (type 4 safety).
|FA141655||Will Zelio Soft 2 V4.5 work on Windows 7 64 bit?|
Yes, the only current issue that arises is when it comes to communicate between the PC/Laptop and the Zelio Logic Relay, if you use the USB cable...
|FA141555||How to check the firmware and hardware version on the Zelio Logic Smart relay?|
Check the firmware on the front of the screen (LCD display) of Zelio Smart Relay.
Check the firmware on the software,...
|FA142621||What address to use for TSXETZ410 and TSXETZ510? Set an address of a TSXETZ410 and TSXETZ510.|
This FAQ is to alert users NOT to set the IP address on any of theses devices of 255.255.255.255.
Do not use this IP address, there is a risk of becoming...
|FA142617||Will Vijeo Designer work on Windows 7 64 bit?|
Yes if the version is V6.0 SP1.1 or higher (like V6.0 SP2.0)
Vijeo Designer will work with Windows 7 32 bit also.
Note: You can NOT install V6.0...
|FA142642||How to connect XPSMC via USB, using TSXCUSB485 USB converter?|
XPSMC + TSXCUSB485 hardware connection
XPSMC + TSXCUSB485 software rules
|FA140674||SR2 and SR3 LCD display screen not working properly (flashing, strange characters, black screen or stripes).What is wrong with it?|
SR2 and SR3
If your Zelio Logic Smart relay screen looks something like this.
If you bought your Zelio Smart relay since June 2011 you will have...
|FA35830||Question Why does the Sepam Series 20, 40 and 80 Relay not display waveforms for all phases of current / voltage? |
When certain variables are calculated rather than acquired (measured) by Sepam, the waveform associated to those variables is not available. Those...
|FA35833||How to retrieve a fault recording (OPG) in SFT2841 from the Sepam Series 20, 40 and 80 Relay? |
Steps to retrieve fault recording in SFT2841:
(Note: Screen Caps may appear differently depending on Sepam series)
1. Verify communication between...